Discover the Cyco Calibration Solution pH 7.0
The most common reasons for pH meter error is dry storage of the electrode and electrode contamination. These cause loss of accuracy, reduced performance and slower response times. Avoid placing pH electrodes in harsh chemicals especially those of high concentration or extreme pH e.g. pH Up. These can damage electrodes or increase the need for electrode maintenance. Thoroughly clean the electrode after use and always store your pH meter with the cap firmly in place. Frequency of meter calibration is crucial and will optimize your pH meter’s performance.
- For calibration of pH Meters at pH 7.0
- pH Buffer 7.0 @ 25° C (77° F)
- Based on international (APHA) standard formula
After your purchase, the items are subject to the respective manufacturer’s warranty policy.
These policies can be found on the original manufacturer’s website or by contacting them.
Indoor Growing Canada is not responsible for replacing, repairing or any costs associated with your item while under warranty.
Indoor Growing Canada will assist you in your warranty claim by helping you coordinate with the original manufacturer.